All rights reserved. Any duplication, adaption or translation without prior written consent is prohibited unless otherwise provided for in the relevant copyright laws.
Initial Release: 3/6/2010
452 Randolph Street, Abington, MA 02351 USA
Toll Free (888) 724-8763 | Ph (781) 982-7000 | Fax (781) 982-7001 | firstname.lastname@example.org
INSTRUCTIONS FOR USE
Emergo Europe Prinsessegracht 20, 2514 AP The Hague, The Netherlands
Become familiar with the features of patient positioning device before use with a patient. Always practice use on a nurse, physician or appropriate volunteer prior to using clinically.
The SpringLoc Clamp is spring-loaded and attaches anywhere along surgical table’s side rail. Accepts 1” x 1/4” flat bar mounting post. Holds all accessories in vertical position.
Attaching to the Side Rail
Detaching from the Side Rail
- Place clamp on rail so flat bottom is in down position
- Push clamp down on rail until springs click
- Secure clamp by turning handle clockwise to locked position
- Unlock clamp by turning handle counterclockwise and remove accessory
- Slide clamp away from standoffs, then push top of clamp down with thumb until it clicks, then grab handle and lift clamp from side rail
Device Dimensions (maximum)
Intended use is to hold operating room accessories on surgical table. Clamp is primarily used together with our Great White Stirrups.
- Length: 3.00” +/- 0.5”
- Width: 2.5” +/- 0.5”
- Depth: 2.25”+/- 0.5”
- Handle Collapsed: 5.5” +/- 0.5”
- Handle Extended: 5.5” +/- 0.5”
- Device Weight: 0.5 +/- 0.5 lbs
- Attaches to surgical table anywhere on rail
- Single-person installation
Overview of Components
SPRINGLOC CLAMP is a surgical table clamp that attaches to surgical table side rails. Clamp is primarily used in conjunction with our Great White Stirrups.
See side rail dimensions below.
US: 0.374” X 1.122” (9.5 mm x 28.5 mm) PN# 800-0038
Denyer: 0.236” x 1.496” (6 mm x 38 mm) PN# 800-0038-DEN
Europe: 0.394” x 0.984” (10 mm x 25 mm) PN# 800-0038-EU
Japan: 0.354” x 1.260” (9 mm x 32mm) PN# 800-0038-JPN
Swiss: 0.394” x 1.1181” (10mm x 30mm) PN# 800-0038-SWISS
- Product not made with (NRL) Natural Rubber Latex
- Device supports 800 lb. proportional patient load (6’4” tall patient per 99% human body model)
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- Product is maintenance-free, check product condition before next use
- Life of device is 5 years under normal use
- Store device between –4 °F to +86°F
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
Product Use Warnings
Maximum load should not exceed appropriate proportion of a patient weighing 800 lbs. Use care with low-maximum load capacity surgical tables that accessory rails are not overloaded. Strictly follow instructions for use for your Operating Table system.
Hazard resulting from incorrect use. Strictly follow the instructions for use for your Operating Table system.
Strictly read/follow manufacturer’s directions for cleaning fluids. DO NOT use cleaners containing phenolics.
Follow current Association of periOperative Registered Nurses (AORN) Journal Guidelines for proper cleaning and disinfection procedure.
- Remove major contaminants from accessory with disposable materials. Follow appropriate bio-hazard waste disposal procedures.
- Apply cleaning fluid liberally to entire accessory and wipe with clean, lint-free cloth until all moisture and cleaning fluid is removed from accessory
- Let accessory dry
IFU-800-0338 Rev 3.00