Download IFU pdf
View/Download Available Translations
CamLoc Rail Clamp 800-0271
CamLoc Rail Clamp, Large 800-0271-L
CamLoc Rail Clamp (DEN) 800-0271-DEN
CamLoc Rail Clamp (EU) 800-0271-EU
CamLoc Rail Clamp (JPN) 800-0271-JPN
CamLoc Rail Clamp (UK) 800-0271-UK
CamLoc Rail Clamp (SWISS) 800-0271-SWISS
Intended use for CamLoc Rail Clamp is to hold accessories to surgical table. The intended users of this device are medical professionals within hospitals and surgery centers.
- Choose Standard clamp (800-0271) to use with 1-2 drapes or Large clamp (800-0271-L) to use with 3-5 drapes
- Product not made with (NRL) Natural Rubber Latex
- Device supports 500 lb. (227 kg) proportional patient load (6’4” (193 cm) tall patient per 99% human body model)
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- CE marked medical device according to MDR (EU) 2017/745
- Product is maintenance-free, check product condition before next use
- Life of device is 5 years under normal use
- Store device between -4°F to +86°F (-20°C to 30°C)
INSTRUCTIONS FOR USE
Place clamp onto rail (as shown). Can be used with 1-2 drapes only.
CamLoc locks to rail from the bottom. Clamp cannot be attached where there is a notch on side rail–it will not be secured or locked.
Lock clamp by turning stainless steel ball clockwise or counterclockwise (as shown).
Inset accessory (as shown).
Device Dimensions (maximum)
Lock accessory by turning star knob to clockwise (as shown).
- Length: 4 3/4” +/- 0.5” (12 cm +/- 1 cm)
- Width: 5” +/- 0.5” (13 cm +/- 1 cm)
- Depth: 2 1/4” +/- 0.5” (6 cm +/- 1 cm)
- Device Weight: 2.0 +/- 0.5 lbs. (.9 kg +/- .22 kg)
- Attaches to rail of surgical table at any point on rail
- Single-person installation
CamLoc Rail Clamp is a surgical table clamp that attaches to side rails of any table.
See side rail dimensions below:
US: 0.374” x 1.122” (9.5 mm x 28.5 mm) PN# 800-0271
Denyer: 0.236” x 1.496” (6 mm x 38 mm) PN# 800-0271-DEN
Europe: 0.394” x 0.984” (10 mm x 25 mm ) PN# 800-0271-EU
Eschmann (UK): 0.236” x 1.260” (6 mm x 32 mm) PN# 800-0271-UK
Japan: 0.354” x 1.260” (9 mm x 32 mm ) PN# 800-0271-JPN
Swiss: 0.394” x 1.181” (10 mm x 30 mm ) PN# 800-0271-SWISS
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
PRODUCT USE WARNINGS
Maximum load should not exceed appropriate proportion of a patient weighing 500 lbs. (227 kg). Use care with low-maximum load capacity surgical tables that accessory rails are not overloaded.
Hazard resulting from incorrect use. Be absolutely sure to follow the instructions for use for your Operating Table system.
Do not reuse device if there are obvious signs of damage or functional issues. Consult manufacturer before reusing.
Clean CamLoc Rail Clamp thoroughly before using & sterilization and after each use.
CLEANING INSTRUCTIONS?Clean CamLoc Clamp thoroughly before using (before sterilization) and after each use
Recommended cleaning products: (All solutions should be used in accordance with manufacturer’s instructions)
- Enzymatic neutral pH detergent solution
- (Enzymatic cleaners are compatible with metal devices/instruments and are preferred)
- Non-enzyme, hydrogen peroxide based solution
- Coverage Plus NPD One Step Cleaner, Disinfectant & Deodorizer Concentrate
- Scrub each piece with a soft, plastic scrub brush and a recommended cleaning solution (see above)
- Rinse all parts thoroughly to remove any chemical residue
- Wipe each part completely dry with a clean, lint-free cloth
- Put through washer/decontaminator according to manufacturer’s instruction (Select cycle that does not included lubrication)
- Prepare for sterilization (device must be cleaned before sterilization)
- See sterilization instructions below
*Based on information provided by the Healthcare Infection Control Practices Advisory Committee (HICPAC), the Center for Disease Control (CDC), the National Center for Biotechnology Information (NCBI) and the U.S. National Library of Medicine.
Wrap using envelope method or accepted wrap method at your facility.
Exposure Time at 132°C (270°F): 4 minutes
Drying Time: 40 minutes
Any serious incident that has occurred in relation to the device should be reported to the manufacturer and the competent authority of the Member State in which the user and/or patient is established.
Drying Time: 40 minutesDrying Time: 40 minutesDrying Time: 40 minutesExposure Time at 132°C (270°F): 4 minutesExposure Time at 132°C (270°F): 4 minutesWrapped InstrumentsWrapped InstrumentsPre-Vacuum?Wrapped Instruments?Exposure Time at 132°C (270°F): 4 minutes?Drying Time: 40 minutes
IFU-800-0271 REV 3.05
Basic UDI-DI: 081005737F0018KY
Latest Revision: 2022-01
eIFU Language Versions
To download and print the Instructions for Use, please click here.
452 Randolph Street, Abington, MA 02351 USA
Toll Free ( 888 ) 724-8763 | Ph (781) 982-7000
Fax (781) 982-7001 | firstname.lastname@example.org
Emergo Europe, Prinsessegracht 20, 2514 AP The Hague, The Netherlands