All rights reserved. Any duplication, adaption or translation without prior written consent is prohibited unless otherwise provided for in relevant copyright laws.
SCHURE SOCKET XL
Initial Release: 5/14/2009
452 Randolph Street, Abington, MA 02351 USA
Toll Free (888) 724-8763 | Ph (781) 982-7000 | Fax (781) 982-7001 | firstname.lastname@example.org
Emergo Europe, Prinsessegracht 20, 2514 AP The Hague, The Netherlands
INSTRUCTIONS FOR USE
Become familiar with patient positioning device’s features before use with patient. Always practice use on a nurse, physician or appropriate volunteer prior to using clinically.
Fits all standard operating room table side rails (3/8”W x 1-1/8”H). Clark socket accepts any round mounting posts up to 5/8” in diameter and rectangular mounting posts up to 1” x 3/8”.
Attaching Clamp to Surgical Table
Detaching Clamp from Surgical Table
- Snap Schure Socket XL to surgical table side rail with logo sticker side up
- Place round or flat post in mounting hole
- To tighten post, turn handle clockwise until locked
- Turn handle counterclockwise until post loosens
- Remove post from socket
- Detach from rail
Intended use is to hold surgical accessories on surgical table.
Device Dimensions (maximum)
- Length: 4” +/- 0.5”
- Width: 2.25” +/- 0.5”
- Depth: 2.25” +/- 0.5”
- Device Weight: 1.5 +/- 0.5 lbs
- Attaches to surgical table rail at any point on rail
- Single-person installation
- Twist lock/release handle
SCHURE SOCKET XL is a surgical table clamp that attaches to side rails of any table.
See side rail dimensions below:
US: 0.374” X 1.122” (9.5 mm x 28.5 mm) PN# 800-0134
Denyer: 0.236” x 1.496” (6 mm x 38 mm) PN# 800-0134-DEN
Europe: 0.394” x 0.984” (10 mm x 25 mm) PN# 800-0134-EU
Eschmann (UK): 0.236” x 1.260” (6mm x 32mm) PN# 800-0134-UK
Japan: 0.354” x 1.260” (9 mm x 32mm) PN# 800-0134-JPN
Swiss: 0.394” x 1.1181” (10mm x 30mm) PN# 800-0134-SWISS
- Product not made with (NRL) Natural Rubber Latex
- Device supports 800 lb. proportional patient load (6’4” tall patient per 99% human body model)
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- Product is maintenance-free, check product condition before next use
- Life of device is 5 years under normal use
- Store device between –4°F to +86°F
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
Product Use Warnings
WARNING! Maximum load should not exceed appropriate proportion of a patient weighing 800 lbs. Use care with low-maximum load capacity surgical tables that accessory rails are not overloaded. Strictly follow instructions for use for your Operating Table system.
WARNING! Hazards result from incorrect use. Strictly follow instructions for use with your Operating Table system.
Follow current Association of periOperative Registered Nurses (AORN) Journal Guidelines for proper cleaning and disinfection procedure.
Strictly read/follow manufacturer’s directions for cleaning fluids. DO NOT use cleaners containing phenolics.
Remove major contaminants from accessory with disposable materials. Follow appropriate bio-hazard waste disposal procedures.
- Apply cleaning fluid liberally to entire accessory and wipe with clean, lint-free cloth until all moisture and cleaning fluid is removed from accessory
- Let accessory dry
Date of Manufacture
Authorized Representative in European Community
Batch Code/Serial Number
IFU-800-0134 REV 3.00