Download IFU pdf
All rights reserved. Any duplication, adaption or translation without prior written consent is prohibited unless otherwise provided in relevant copyright laws.
Lithotomy Stirrups 800-0049
Initial Release: 3/10/2007
452 Randolph Street, Abington, MA 02351 USA
Toll Free (888) 724-8763 | Ph (781) 982-7000
Fax (781) 982-7001 | email@example.com
Emergo Europe, Prinsessegracht 20, 2514 AP The Hague, The Netherlands
INSTRUCTIONS FOR USE
Become familiar with the features of patient positioning device before use with a patient. Always practice use on a nurse, physician or appropriate volunteer prior to using clinically.
Attaching Stirrups to Surgical Table
- Attach Schure Socket Clamps (sold separately) onto table side rails.
- Insert Lithotomy Stirrup mounting posts into clamps. Turn Schure Socket handle clockwise to tighten.
- Put patient’s legs into boots and fasten with loop and hook straps.
- Turn boot block handles counterclockwise to loosen boot blocks. Adjust up or down to desired position. Turn boot block handle clockwise to tighten.
Detaching Stirrups from Table
- Turn Schure Socket handle counterclockwise to loosen and pull stirrup from socket
Intended use is to hold patient’s legs during short or long gynecologic, urological or laparoscopic procedures.
Device Dimensions (maximum)
- Length 26.5” +/- 0.5”
- Width: 6” +/- 0.5”
- Depth: 2.5” +/- 0.5” (with pad)
- Device Weight: 5.25 +/- 0.5 lbs.
- Attaches to rail of surgical table at any point on the rail
- Single-person installation
- Push Release Trigger
LITHOTOMY STIRRUPs are a surgical table accessory that aids in positioning of legs for gynecological, urological, or laparoscopic procedures.
Other required products for use:
800-0006 Schure Socket
, 2 each (sold separately)
US: 0.374” x 1.122” (9.5 mm x 28.5 mm) PN# 800-0006
Denyer: 0.236” x 1.496” (6 mm x 38 mm) PN# 800-0006 -DEN
Europe: 0.394” x 0.984” (10 mm x 25 mm ) PN# 800-0006 -EU
Eschmann (UK): 0.236” x 1.260” (6 mm x 32 mm) PN# 800-0006-UK
Japan: 0.354” x 1.260” (9 mm x 32 mm ) PN# 800-0006-JPN
Swiss: 0.394” x 1.181” (10 mm x 30 mm ) PN# 800-0006-SWISS
- Product not made with (NRL) Natural Rubber Latex
- Device supports 350 lb. proportional patient load (6'4" tall patient per 99% human body model)
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- Product is maintenance-free, check product condition before next use
- Life of device is 5 years under normal use
- Store device between –4°F to +86°F
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
Product Use Warnings
Maximum load should not exceed appropriate proportion of a patient weighing 350 lbs. Use care with low-maximum load capacity surgical tables that accessory rails are not overloaded.
Hazard resulting from incorrect use. Strictly follow Instructions for Use with your Operating Table system.
Surgical table load capacities may be less. Never overload a surgical table. Device is intended for mounting on side of surgical table rail only.
Follow current Association of periOperative Registered Nurses (AORN) Journal Guidelines for proper cleaning and disinfection procedure.
Adhere to standards for blood-borne pathogens from the Occupational Safety and Health Administration. Use recommended protective clothing, gloves, masks and eye protection to clean accessory.
Strictly read/follow manufacturer’s directions for cleaning fluids. DO NOT use cleaners containing phenolics.
- Remove major contaminants from accessory with disposable materials. Follow appropriate bio-hazard waste disposal procedures.
- Apply cleaning fluid liberally to entire accessory and wipe with clean, lint-free cloth until all moisture and cleaning fluid is removed from accessory
- Let accessory dry
Date of Manufacture
Authorized Representative in European Community
Batch Code/Serial Number
IFU-800-0049 REV 3.01