Download IFU pdf
All rights reserved. Any duplication, adaption or translation without prior written consent is prohibited unless otherwise provided for in relevant copyright laws.
Leg Prepper 800-0013
Leg Prepper Pad and Cover 790-0074
Initial Release: 12/2/2011
452 Randolph Street, Abington, MA 02351 USA
Toll Free (888) 724-8763 | Ph (781) 982-7000
Fax (781) 982-7001 | email@example.com
INSTRUCTIONS FOR USE
Emergo Europe, Prinsessegracht 20
2514 AP The Hague, The Netherlands
Become familiar with patient positioning device’s features before use with patient. Always practice use on a nurse, physician or appropriate volunteer prior to using clinically.
Attaching Leg Prepper to Table
Detaching Leg Prepper from Surgical Table
- Snap Schure Socket XL P/N 800-0134 (sold separately) onto the rail
- Insert mounting post into socket - lock tight
- Turn clamp handle counterclockwise to loosen Leg Prepper and remove
Intended use is to hold patient’s leg comfortably during prepping procedure.
Device Dimensions (maximum)
- Length: 36.5” +/- 0.5”
- Width: 12” +/- 0.5”
- Depth: 2” +/- 0.5”
- Device Weight: 4 +/- 0.5 lbs.
- 5/8” diameter mounting post inserts into socket
- Single-person installation
- Twist lock/Release Handle
LEG PREPPER is a surgical table accessory. Padded horseshoe design holds patient’s leg comfortably during prepping procedure.
Leg Prepper Pad And Cover
Other required product for use:
800-0134 Schure Socket XL (sold separately)
US: 0.374” x 1.122” (9.5 mm x 28.5 mm) PN# 800-0134
Denyer: 0.236” x 1.496” (6 mm x 38 mm) PN# 800-0134-DEN
Europe: 0.394” x 0.984” (10 mm x 25 mm ) PN# 800-0134-EU
Eschmann (UK): 0.236” x 1.260” (6 mm x 32 mm) PN# 800-0134-UK
Japan: 0.354” x 1.260” (9 mm x 32 mm ) PN# 800-0134-JPN
Swiss: 0.394” x 1.181” (10 mm x 30 mm ) PN# 800-0134-SWISS
- Product not made with (NRL) Natural Rubber Latex
- Device supports 500 lb. proportional patient load (6'4" tall patient per 99% human body model)
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- Product is maintenance-free, check product condition before next use
- Life of device is 5 years under normal use
- Store device between –4°F to +86°F
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
Product Use Warnings
Maximum load should not exceed appropriate proportion of a patient weighing 500 lbs. Use care with low-maximum load capacity surgical tables that accessory rails are not overloaded.
Hazards result from incorrect use. Strictly follow Instructions for Use for your Operating Table system.
Surgical table load capacities may be less. Never overload a surgical table. Device is intended for mounting on side of surgical table rail only.
Follow current Association of periOperative Registered Nurses (AORN) Journal Guidelines for proper cleaning and disinfection procedures.
Adhere to standards for blood-borne pathogens from Occupational Safety and Health Administration (OSHA). Use recommended protective clothing, gloves, masks and eye protection to clean accessory.
Strictly read/follow manufacturer’s directions for cleaning fluids. DO NOT use cleaners containing phenolics.
- Remove major contaminants from accessory with disposable materials. Follow appropriate bio-hazard waste disposal procedures.
- Apply cleaning fluid liberally to entire accessory and wipe with clean, lint-free cloth until all moisture and cleaning fluid is removed from accessory
- Let accessory dry
Date of Manufacture
Authorized Representative in European Community
Batch Code/Serial Number
IFU-800-0013 REV 3.00