Download IFU pdf
All rights reserved. Any duplication, adaption or translation without prior written consent is prohibited unless otherwise provided for in the relevant copyright laws.
Anesthesia Screen 800-0016
Initial Release: 3/15/2015
452 Randolph Street, Abington, MA 02351 USA
Toll Free (888) 724-8763 | Ph (781) 982-7000
Fax (781) 982-7001 | email@example.com
INSTRUCTIONS FOR USE
Emergo Europe, Prinsessegracht 20, 2514 AP The Hague, The Netherlands
Become familiar with the features of patient positioning device before use with a patient. Always practice use on a nurse, physician or appropriate volunteer prior to using clinically.
Attaching to Side Rail
Detaching from Side Rail
- Attach Schure Socket XL (sold separately) on rail. Insert mounting post into socket and lock into position by turning handle clockwise.
- Adjust screen to desired height. It is adjusts from 15” to 33”. Lock in desired position by turning height adjustable lock handle clockwise.
- Unlock clamp by turning handle counterclockwise and remove mounting post from socket
Intended use is to aid anesthesiologist in viewing patient during surgery.
Device Dimensions (maximum)
- Length: 23” +/- 0.5”
- Height: 35 3/8” +/- 0.5”
- Depth: 1 3/4”+/- 0.5”
- Device Weight: 3 +/- 0.5 lbs.
- Attaches to rail of surgical table at any point on rail
- Single-person installation
ANESTHESIA SCREEN is a surgical table accessory that aids anesthesiologist in viewing patient during surgery.
Required product for use: 800-0134 Schure Socket XL (sold separately)
US: 0.374” x 1.122” (9.5 mm x 28.5 mm) PN# 800-0134
Denyer: 0.236” x 1.496” (6 mm x 38 mm) PN# 800-0134-DEN
Europe: 0.394” x 0.984” (10 mm x 25 mm ) PN# 800-0134-EU
Eschmann (UK): 0.236” x 1.260” (6 mm x 32 mm) PN# 800-0134-UK
Japan: 0.354” x 1.260” (9 mm x 32 mm ) PN# 800-0134-JPN
Swiss: 0.394” x 1.181” (10 mm x 30 mm ) PN# 800-0134-SWISS
- Product not made with (NRL) Natural Rubber Latex
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- Product is maintenance-free, check product condition before next use
- Life of device is 5 years under normal use
- Store device between –4°F to +86°F
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
Follow current Association of periOperative Registered Nurses (AORN) Journal Guidelines for proper cleaning and disinfection procedures.
Adhere to standards for blood-borne pathogens from the Occupational Safety and Health Administration (OSHA). Use recommended protective clothing, gloves, masks and eye protection to clean accessory.
Strictly read/follow manufacturer’s directions for cleaning fluids. DO NOT use cleaners containing phenolics.
- Remove major contaminants from accessory with disposable materials. Follow appropriate bio-hazard waste disposal procedures.
- Apply cleaning fluid liberally to entire accessory and wipe with clean, lint-free cloth until all moisture and cleaning fluid is removed from accessory
- Let accessory dry
Date of Manufacture
Authorized Representative in European Community
Batch Code/Serial Number
IFU-800-0016 REV 3.01