Download IFU pdf
All rights reserved. Any duplication, adaption or translation without prior written consent is prohibited unless otherwise provided in relevant copyright laws.
Lateral Shoulder Holder 800-0169
Disposable Sterile Hand/Forearm Wrap, 10/cs 800-0281-F
Initial Release: 1/22/15
452 Randolph Street, Abington, MA 02351 USA
Toll Free (888) 724-8763 | Ph (781) 982-7000
Fax (781) 982-7001 | firstname.lastname@example.org
Emergo Europe, Prinsessegracht 20, 2514 AP The Hague, The Netherlands
(1) 700-0512 • (2) 790-0011 • (3) 790-0013
(4) 790-0014 • (5) 790-0015 • (6) 800-0281-F
INSTRUCTIONS FOR USE
Become familiar with patient positioning device’s features before use with patient. Always practice on a nurse, physician or appropriate volunteer prior to using clinically.
Attaching Lateral Shoulder Holder to Surgical Table
- Attach Camloc Clamp (sold separately), over drape, anywhere along surgical table side rail
- Insert 790-0011 into clamp and tighten handle to lock into place
- Loosen lever and insert 790-0013 into 790-0011. Adjust to desired height and lock lever tight.
- Loosen handle and pull plunger to slide 790-0014 into 790-0013. Rotate 790-0014 to desired position, lock handle and release plunger.
- Insert 790-0015 into appropriate slot for desired position, listen for click
Placing and Securing the Patients Arm
With the Disposable Sterile Hand/Forearm Wrap (800-0281-F)
- Place patient arm into Forearm Pad, ensuring patient’s fingers wrap around Forearm Attachment Handle and wrap pad snugly around patient's arm. See Figures 1-2.
- Using self-adherent wrap, secure patients arm to forearm attachment by wrapping entire assembly. Be sure to secure hands and fingers with wrap. See Figures 3-4.
Instructions for Use of the Lateral Shoulder Holder
Removing Lateral Shoulder Holder
- Insert 700-0512 into appropriate slot of 790-0015. The plunger will pop into place.
- Loosen handle to allow rotation of arm. Tighten handle when arm has reached desired position.
- For traction turn knob right to tighten and left to loosen
- Pull plunger to remove arm from device. Unwrap arm.
- Release plunger to remove traction system (790-0015)
- Pull plunger and loosen handle to release (790-0014)
- Loosen lever to remove pole (790-0013)
- Loosen clamp and remove (790-0011)
- Follow cleaning and sterilization instructions below
Intended use is to provide traction and rotation of operative shoulder controlled by surgeons preference, during shoulder surgery.
Device Dimensions (maximum)
- Length: 30” +/- 0.5” (extended), 9” +/- 0.5” (collapsed)
- Height: 48” +/- 0.5” (extended), 29”+/- 0.5” (collapsed)
- Net weight with clamp 13.8 +/- 0.5 lbs.
- Net weight without clamp 12.5 +/- 0.5 lbs.
- Gross Weight 19.5 +/- 0.5 lbs.
- Range of Motion: 360° Rotation
- Single-person installation
Overview of Components
LATERAL SHOULDER HOLDER is a device used during shoulder surgery. Its function is to provide traction and rotation of the operative shoulder controlled by the surgeons preference, during shoulder surgery.
A CamLoc Clamp (sold separately) is required to attach Lateral Shoulder Holder to surgical table and Disposable Sterile Hand/Forearm Wrap that attaches the patient limb to the Lateral Shoulder Holder.
- US: 0.374” x 1.122” (9.5 mm x 28.5 mm) PN# 800-0271
- Denyer: 0.236” x 1.496” (6 mm x 38 mm) PN# 800-0271-DEN
- Europe: 0.394” x 0.984” (10 mm x 25 mm ) PN# 800-0271-EU
- Eschmann (UK): 0.236” x 1.260” (6 mm x 32 mm) PN# 800-0271-UK
- Japan: 0.354” x 1.260” (9 mm x 32 mm ) PN# 800-0271-JPN
- Swiss: 0.394” x 1.181” (10 mm x 30 mm ) PN# 800-0271-SWISS
800-0281-F Disposable Sterile Hand/Forearm Wrap
- Product not made with (NRL) Natural Rubber Latex
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- Life of device is 3 years under normal use
- Store device between –4°F to +86°F
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
Product Use Warnings
WARNING! Maximum load should not exceed appropriate proportion of a patient weighing 500 lbs. Use care with low-maximum load capacity surgical tables that accessory rails are not overloaded.
WARNING! Hazards result from incorrect use. Strictly follow Instructions for Use with your Operating Table system.
Cleaning and Sterilization Instructions
Lateral Shoulder Holder is used to support the arm of the patient. No other additional forces, may act on the Lateral Shoulder Holder.
- Clean TKR Positioner thoroughly before using (before sterilization) and after each use
- Recommended cleaning products: (All solutions should be used in accordance with manufacturer’s instructions)
- Enzymatic neutral pH detergent solution
- (Enzymatic cleaners are compatible with metal devices/instruments and are preferred)
- Non-enzyme, hydrogen peroxide based solution
- Coverage Plus NPD One Step Cleaner, Disinfectant & Deodorizer Concentrate
- Disassemble TKR Positioner into separate parts for cleaning:
- Boot Ball
- Mounting Plate with Clamp
- Base Plate with Track
- Slide Block with Handle
- Scrub each piece with a soft, plastic scrub brush and a recommended cleaning solution (see above)
- Rinse all parts thoroughly to remove any chemical residue
- Wipe each part completely dry with a clean, lint-free cloth
- Put through washer/decontaminator according to manufacturer’s instruction (Select cycle that does not included lubrication)
- Prepare for sterilization (device must be cleaned before sterilization)
- See sterilization instructions below
*Based on information provided by the Healthcare Infection Control Practices Advisory Committee (HICPAC), the Center for Disease Control (CDC), the National Center for Biotechnology Information (NCBI) and the U.S. National Library of Medicine.
Exposure Time at 132°C (270°F): 4 minutes
Drying Time: 40 minutes
Date of Manufacture
Authorized Representative in European Community
Batch Code/Serial Number
IFU-800-0169 REV 3.04