Download IFU pdf
All rights reserved. Any duplication, adaption or translation without prior written consent is prohibited unless otherwise provided in relevant copyright laws.
Great White Robotic Stirrups 800-0342-R
Great White Robotic Stirrups with Fins 800-0342-RF
Replacement Boot Pads 508-1354
Replacement Boot Pads with Fins 508-1415
Initial Release: 1/25/2017
452 Randolph Street, Abington, MA 02351 USA
Toll Free (888) 724-8763 | Ph (781) 982-7000
Fax (781) 982-7001 | email@example.com
Emergo Europe, Prinsessegracht 20, 2514 AP The Hague, The Netherlands
INSTRUCTIONS FOR USE
Become familiar with patient positioning device’s features before use with patient. Always practice on a nurse, physician or appropriate volunteer prior to using clinically.
- Attach SpringLoc Clamps P/N 800-0338 on accessory rails in same location on opposite sides of surgical table at patient’s hip joints.
- Prior to placing device into SpringLoc Clamp, identify patient’s left and right side stirrup indicated on boot label.
- Insert stirrup blades into clamps. Tighten clamps by turning handle to the right.
- Ensure patient is positioned on surgical table in accordance with procedure and surgeon requirements.
WARNING!! To prevent patient or operator injury from inadvertent stirrup movement, securely tighten accessory clamp and boot clamp.
(1 & 4) Free Floating Boot Pivot, (2) Boot Clamp, (3) Trigger Handle
- Support patient’s leg by grasping heel in one hand and underside of knee with other hand. Gently flex knees and transfer leg into boot, then secure boot straps.
To achieve appropriate leg/foot position, loosen boot clamp. Adjust boot to desired position and re-tighten clamp. Ensure patient’s heels are securely seated in boot heels.
To achieve appropriate lithotomy and abduction positions, squeeze trigger, adjust to desired position and release to lock.
Note: Free-floating boot is designed to rotate about pivot shown above through-out the full range of motion, reducing risk of superficial nerve injury.
- Loosen clamps and remove stirrups by lifting them out of clamps
- Remove accessory clamps from side rails
Device Dimensions (maximum)
Great White Robotic Stirrups are a patient positioning accessory used in Gynecology, Urology, and laparoscopic procedures. They provide a safe system for positioning the legs up to 600 lb. patients in a variety of lithotomy/abduction positions. These are the only stirrups that have been designed specifically for use in robotic assisted procedures. Stirrups can be lowered approximately 25° lower than conventional stirrups.
- Height: 35.25” +/- 0.5”
- Width: 9” +/- 0.5”
- Depth: 14” +/- 0.5”
- Device Weight: 10 +/- 0.5 lbs (with pad)
- Stirrups have a range of motion -60° to 40° in the lithotomy position and -9° abduction to 25° abduction.
- Single-person installation
- Connects to surgical table seat section
- Stirrups store on transport/storage dolly (P/N 800-0074-R sold separately)
Great White Robotic Stirrups and Great White Robotic Stirrups with Fins provide a safe system for positioning the legs up to 600 lb. patients in a variety of lithotomy/abduction positions.
800-0342-R & 800-0342-RF
Other required products for use:
800-0338 SpringLoc Clamp
800-0338-EU SpringLoc Clamp
800-0338-JPN SpringLoc Clamp
800-0338-SWISS SpringLoc Clamp
508-1415 Replacement Boot Pads with Fins
508-1354 Replacement Boot Pads without Fins
- Product not made with (NRL) Natural Rubber Latex
- Device supports 600 lb. proportional patient load (6'4" tall patient per 99% human body model)
- Product warranty covers product from manufacturing defects for period of 2 years
- If damaged in shipping, call Customer Service at (888) 724-8763 or (781) 982-7000 to obtain Return Material Authorization (RMA) number. For product warranty issues, contact Customer Service.
- Life of device is 5 years under normal use
- Store device between –4 °F to +86 °F
- General - Prevent infection by cleaning and disinfecting product before disposal
- Packaging - Dispose packaging material via household waste according to national requirements
- SchureMed accepts back used or retired products - or dispose of product in accordance with national requirements
Product Use Warnings
Maximum load should not exceed appropriate proportion of a patient weighing 600 lbs. Use care with low-maximum load capacity surgical tables that accessory rails are not overloaded.
Hazard resulting from incorrect use. Strictly follow instructions for your Operating Table system.
Surgical table load capacities may be less. Never overload a surgical table. The device is intended for mounting on the side of a rail of a surgical table only.
Follow current Association of periOperative Registered Nurses (AORN) Journal Guidelines for proper cleaning and disinfection procedure.
Adhere to standards for blood-borne pathogens from the Occupational Safety and Health Administration (OSHA). Use recommended protective clothing, gloves, masks and eye protection to clean accessory.
Strictly read/follow manufacturer’s directions for cleaning fluids. DO NOT use cleaners containing phenolics.
- Remove major contaminants from accessory with disposable materials. Follow appropriate bio-hazard waste disposal procedures.
- Apply cleaning fluid liberally to entire accessory and wipe with clean, lint-free cloth until all moisture and cleaning fluid is removed from accessory
- Let accessory dry
Date of Manufacture
Authorized Representative in European Community
Batch Code/Serial Number
IFU-800-0342-R-RF REV 3.00